|Statement||[by] Malcolm Dole.|
|Contributions||Coleman Electric Co. (Maywood, Ill.)|
|LC Classifications||QD561 .D67|
|The Physical Object|
|Number of Pages||20|
|LC Control Number||42045697|
There are various methods of measuring pH, but the main methods are those using chemical indicators, hydrogen electrodes, antimony electrodes, or glass electrode. Here glass electrode method is described. pH measurements can be most accurately made with a pH measurement “loop” consisting of a probe, a preamplifier, and an analyzer. made of a simple glass-electrode assembly for measuring the pH value of leather extracts and tannin solutions. A method is proposed and the details of operation noted which are essential for accurate routine pH measurements. The Kerridge method of converting the emf readings of the glass electrode to corresponding pH units is described. CONTENTSFile Size: 7MB. through the process line damage the glass electrode. To prevent vibrations in the process from damaging the sensor, pH sen-sors may incorporate technology to transfer damaging energy away from the rigid glass-to-metal seal within the glass electrode body. Accurate Calibration The best way to verify the accuracy of a pH analyzer and sensor is by. activity, or rather, its negative logarithm, the pH, is so easy to measure, pH is the most commonly monitored and recorded parameter of liquid samples. A glass electrode (Fig. 1) is actually a device, not an elec-trode in an electrochemical sense of the word. It consists of a glass bulb membrane, which gives it its name and an elec-.
The pH electrodes are available with a glass or epoxy body and a variety of electrode body styles, fill types, references, junctions, and connectors that suit your application. To learn more, see our pH electrodes separated by category and application below. Among various potentiometric pH sensors, conventional glass pH electrodes are widely used and the pH value measured using a glass electrode is often considered as a gold standard in the development and calibration of other novel pH sensors in vivo and in vitro . Other pH electrodes, such as metal/metal oxide and ISFETs have received more and more attention in recent years due to their . The glass electrode for pH Measurement. Glass electrode has become the most useful and convenient electrode in determining the pH of solutions, particularly with the commercially built pH meters. Most often used pH electrodes are glass electrodes. A typical model is made of glass tube ended with small glass . This thin membrane is called the electrode membrane. Normally, when the temperature of the solution is 30 ℃, if the pH inside is different from that of outside by 1, it will create approximately 60 mV of electromotive force. The liquid inside the glass electrode usually has a pH of 7.